Implantable, wireless device platforms for neuroscience research Philipp Gutruf and John A Rogers. NORTH WESTERN UNIVERSITY
Available online at http://www.sciencedirect.com ScienceDirect
Implantable, wireless device platforms for neuroscience research
Philipp Gutruf and John A Rogers
Recently developed classes of ultraminiaturized wireless devices provide powerful capabilities in neuroscience research, as implantable light sources for simulation/inhibition via optogenetics, as integrated microfluidic systems for programmed pharmacological delivery and as multimodal sensors for physiological measurements. These platforms leverage basic advances in biocompatible materials, semiconductor device designs and systems engineering concepts to afford modes of operation that are qualitatively distinct from those of conventional approaches that tether animals to external hardware by means of optical fibers, electrical cables and/or fluidic tubing. Neuroscience studies that exploit the unique features of these technologies enable insights into neural function through targeted stimulation, inhibition and recording, with spatially and genetically precise manipulation of neural circuit activity. Experimental possibilities include studies in naturalistic, three dimensional environments, investigations of pair-wise or group related social interactions and many other scenarios of interest that
cannot be addressed using traditional hardware.
Departments of Materials Science and Engineering, Biomedical Engineering, Chemistry, Neurological Surgery, Mechanical Engineering, Electrical Engineering and Computer Science, Simpson Querrey Institute & Feinberg Medical School, 2145 Sheridan Road, Evanston, IL 60208, United States
Corresponding author: Rogers, John A (email@example.com)
light sensitive ion channels [1,2] and photometry via fluorescent indicators of calcium [3–5] represent recent breakthrough developments of critical importance in this context. The insights already gained by implementation of these methodologies with conventional technologies are numerous; further progress will be accelerated by the development of advanced technologies for delivering optical and pharmacological stimuli and for recording relevant responses .
Conventional means for delivering and/or capturing light from targeted regions in behaving animals use optical fibers inserted with stereotactic apparatus and held in place via surgical glues, cements, sutures and external fixtures [7–9]. Pharmacological delivery and electrical recording occur in conceptually similar modes, but where plastic tubes and metal wires, respectively, replace the optical fibers . The stiff, high moduli materials used for these systems are, however, fundamentally mismatched with the soft, com- pliant tissues of the brain and peripheral nerves. Micro- motions induced by forces delivered through the associ- ated hardware [11,12] together with those that arise from natural, biological movements can lead to degradation at the biotic/abiotic interface through cellular damage and glial scar formation . The physical tethers as well as heavy battery powered wireless hardware also impose strict limitations on behavioral tests , as they prevent experi- ments that involve social interactions or activities in natu- ralistic environments with three dimensional obstacles. In addition, the tethers unavoidably affect the animals by limiting their motion and altering their behaviors in ways that can be difficult to quantify, thereby frustrating exper- imental reproducibility and clouding interpretation of data collected in certain classes of studies.
Emerging technologies enabled by interdisciplinary advances in materials science, optoelectronic devices, system engineering and mechanics design offer some compelling options in this context. Some key system attributes appear in Figure 1a. In all cases, an important goal is for miniaturized, fully implantable platforms with wireless capabilities and battery-free operation, to avoid the limitations that characterize conventional approaches, as outlined above . Biocompatibility at the level of both the chemistry and the mechanics is essential to allow for stable, minimally invasive function within the chal- lenging set of conditions that exist inside the animal and around its environment . Of critical importance for the former is mechanical compliance with and conformal contact to the targeted tissues as the basis of a stable
Current Opinion in Neurobiology 2018, 50:42–49
This review comes from a themed issue on Neurotechnologies Edited by Polina Anikeeva and Liqun Luo
0959-4388/ã 2017 Elsevier Ltd. All rights reserved.
Uncovering the foundational working principles of the brain and the peripheral nervous system represent scien- tific grand challenges, with deep intellectual significance in academic research as well as profound practical value in human healthcare. Here, the advent of genetically tar- geted neuroscience techniques such as optogenetics via
Current Opinion in Neurobiology 2018, 50:1–8
(a) Schematic illustration of key considerations in technologies for neuroscience research. Examples of emerging tools. (b) Multifunctional fiber with electrical, optical and fluidic interfaces . (c) Silicon shank with electrical and optical interfaces . (d) Exploded-view images of a flexible probe with electrical, optical (emission and detection) thermal interfaces . (e) Soft probe with optical and fluidic interfaces .
Mechanical and Mechanical conformality Chemical Durability
Current Opinion in Neurobiology
bio-interface . Highly mobile, time-dynamic tissues intrinsic to many parts of the peripheral nervous system  give rise to difficulties in integration that are even more daunting than those associated with the brain. In all cases, minimizing size and weight is an essential engi- neering consideration .
Injectable, multifunctional platforms
Recent engineering developments greatly expand the scope of capabilities in neural interfaces. Platforms based on optical fibers as simple waveguides to transport light for optical stimulation  have evolved to multimodal probes for delivery of light, fluid and electrical current [20–23], sometimes with in situ recording and computing capabili- ties [24,25]. Other notable advances include syringe injectable electronics that unfold in target tissue to record electrical signals , ultra-thin organic transistor arrays for in situ myoelectric signal recording of optogenetically evoked spikes in muscles [27,28] and multifunctional electrodes for recordings in non-human primates  and wireless ultrasonic powered electrical devices for elec- tromyogram and electroneurogram measurements .
Successful animal behavioral experiments have been reported using multicore fibers shown in Figure 1b in which a core waveguide is surrounded by fluidic channels
and recording electrodes . Another example is in Figure 1c, where optical stimulation occurs via a light emitting diode (LED) monolithically integrated on a silicon platform that also supports electrodes for electrical recordings in the style of a conventional Michigan probe . Traditional wireless hardware has some utility in such cases, but major disadvantages follow from complex con- nections, undesirable mechanics, large sizes and weights [25,33], resulting in requirements for externally mounted head stages that can be cumbersome for small animal models. Monolithically integrated structures that combine diverse classes of materials in optimized mechanics layouts represent qualitatively differentiated types of technology. Here, entire optoelectronic systems with cellular-scale components can be configured as thin, flexible, injectable probes. Figure 1d shows such a miniaturized collection of devices, including a precision temperature sensor to mon- itor thermal load, microscale inorganic LEDs (m-ILEDs) to enable optical stimulation, a micro photodiode to allow for photometry and a micro electrode to perform electrical recording. The entire system has lateral dimensions of 300 mm and uses a thin (20 mm) polyimide support. The cross sectional size and bending stiffness are significantly smaller than those of optical fibers . Another example is in Figure 1e, where a collection of soft, microfluidic channels couple with m-ILEDs to allow combined
Current Opinion in Neurobiology 2018, 50:1–8
pharmacological delivery (up to four different liquid agents) and optogenetic illumination with a single device . Such platforms follow from advanced concepts for materials growth and epitaxial liftoff , lithographic device processing for miniaturized, high-performance components, deterministic assembly methods for hetero- geneous integration , and hard/soft mechanics designs for flexible/stretchable properties . Efficient heat dis- sipation from m-ILEDs, which follows from their large surface area to volume ratios and small thermal mass, is critically important for their safe operation in the brain or adjacent to peripheral nerves .
Strategies in power supply
In nearly all cases, means for power supply are among the most challenging aspects for implementing these types of systems in small animal models. Approaches based on electrochemical energy storage, such as batteries and supercapacitors, consume significant space, they are heavy and often involve toxic materials, and they offer limited operational lifetimes . Reported examples include devices that incorporate signal processing capabilities  as well as high data rate wireless transmission sub- systems  and thermally initiated microfluidic pumps , all supported by externally attached electronics and battery packs . For prolonged operation, detachable modules (size > 1.5 cm2 and weight > 1 g)  allow the battery to be removed for recharging. Particularly for small animal models, weight becomes a primary concern, where even 1 g can be problematic for mice.
Wireless power transfer schemes implemented with min- iaturized circuitry and specialized antenna designs can circumvent these limitations. Here, sufficient power can be generated to operate m-ILEDs, microfluidic pumps and other demanding components, indefinitely through- out regions of interest in home cages (typically 2030cm for mice and 2540cm for rats) and other experimental assays . Table 1 compares features of recently reported approaches. Far field energy transfer
involves electromagnetic waves, typically with frequencies between 420 MHz and 2.4 GHz , that leave the con- finement of the transmission antenna to enable power delivery across large distances, with operation that can be enhanced by use of real-time tracking systems . Disadvantages include efficiencies that depend strongly on relative orientations of the antenna pairs, interferences and standing waves that occur due to reflections from environ- mental obstacles and absorption associated with metal components, water or moisture. The latter effects can be detrimental to systems that are fully implanted into the body  and they limit practical operating ranges to 0.1–3 m. Electromagnetic absorption results in heating of the tissue, thereby limiting the total power that can be used without adverse consequences , representative specific absorption rate (SAR) values for far field powered devices can be found in Table 1. Resulting SAR strongly varies with system efficiency which is specifically problematic for miniaturized devices . To circumvent these disadvan- tages, hybrid solutions that combine far field RF power transfer with solar cells and/or local energy storage can be attractive . Other strategies use containment of the RF field operated at 1.5 GHz in a machined aluminum reso- nant cavity designed to selectively couple the field into the test subject. The result offers a self-tracking effect that reduces the field distribution fluctuations and ambient absorption of traditional far field systems, but is only efficient in close proximity with the cavity .
An alternative uses magnetic resonant coupling [51,52]. Here, a loop antenna wrapped around the test area forms an oscillating magnetic field that can be harnessed by miniaturized receiver coils . In this way, transfer of significant power ($5–10 mW) across cage sizes relevant for small animal models allow miniaturized device dimen- sions with supporting RF transmission hardware that is both compact and efficient. In comparison to far field systems, the same power can be harvested with a reduction in overall device size and SAR  (Table 1) and greatly reduced dependence on orientation and
Implantable, wireless device platforms for neuroscience research Gutruf and Rogers 3
Summary of options in power supply
Weight Embodiment Available power Power source setup
Battery powered [35,39,41]
Recharge needed for continuous operation Depending on wireless communication method ❤ m
Battery free Far field — 1– 2.7 GHz [45,50,55]
Complex primary antenna setup
High (SAR 6150 -69  mW/kg)
Battery free mixed — Solar + Far field 
Complex antenna and light setup
Medium (SAR $20 mW/kg )
Battery free near field resonant magnetic coupling 13.56 MHz 
Simple, versatile primary antenna setup
Low (SAR$ <20 mW/kg )
Current Opinion in Neurobiology 2018, 50:1–8
reflection. For example, a typical miniaturized single channel far field system might occupy an area of 9– 10 mm2 , dominated by the antenna, while mag- netic resonant coupling can be supported using a coil with an area of only 1–2 mm2 . Additionally, the induc- tance of the coil defines the device geometry in a way that yields operation independent of dielectric environment to enable a wide range of design options without the need for extensive optimization of antenna configuration .
Wireless systems for optogenetic studies of the brain
The most recent generation of wireless devices for opto- genetics represents a significant advance over prior work in terms of form factor, weight, biocompatibility and chronic implantability. Optimized antenna designs in the RF regime allow for efficient transfer of energy to power highly miniaturized light sources , in platforms that can be subdermally implanted. As illustrated in Figure 2a, a typical device can be injected into the brain of a small animal model with minimal impact on the host due to small size (5.4 4.3 mm with thickness of 0.7 mm) biocompati- ble encapsulation and stretchable implantable filament with biodegradable polymer support for mechanical sup- port during surgery (width, length and thickness of 0.5 mm, 3.5 mm and 0.1 mm). The device dimensions of the harvester sub-system allow its complete subdermal implantation, thereby resulting in minimal effects on the animal after recovery. Advanced platforms of this type also feature multimodal operation as shown in Figure 2a . Here, two distinct RF frequencies (2.3 and 2.7 GHz) allow for independent control over blue and green light sources, were chosen to be spectrally separated to for independent optogenetic stimulation and inhibition .
Related embodiments provide means to illuminate spa- tially distinct sites for advanced interrogation of neural circuits. The miniaturized antenna uses in this case parallel capacitive coupling of adjacent serpentine wires in the overall layout, thereby reducing the antenna area by 100-fold compared to traditional designs. Experimen- tal demonstrations show robust place aversion and pref- erence with spatially and optically separated stimulation of the ventral and dorsal regions of the nucleus accum- bens shell. Figure 2b shows a similarly miniaturized device (9.8 mm in diameter and maximum thickness of 1.3 mm, with injectable filament of maximum thickness 75 mm) . As in the previous case, complete implan- tation is possible, with stable operation for up to 1-year. Successful biological studies include real-time place pref- erence in animals that express dopaminergic terminals in the nucleus accumbens. The commercial availability of m-ILEDs for these purposes with various emission wave- lengths enables use of this platform with most opsins  and the near field magnetic coupling based power transfer facilitates easy experimental setup with a variety of test enclosures.
A different, but related device formed using hand-wound coils exploits resonant cavity designs, as described previ- ously, to supply high field densities as displayed in Figure 2c . The use of such devices is limited to custom-designed experimental areas that can be difficult to adapt for varied experimental paradigms. Published demonstrations include stimulation of the motor cortex via optical illumination of the right premotor cortex in mice expressing Thy1-ChR2-EYFP to yield an increase in circling locomotion compared to wild type control animals.
Another complementary capability in stimulation and analysis of neural circuitry involves delivery of liquid drugs  in a time locked and highly localized fashion, without tethers and combined with options in optoge- netic activation and viral vector delivery. Figure 2d shows a wireless device for simultaneous, and independently controlled, optical and fluidic delivery (up to four differ- ent drugs, separately) with demonstration of drug induced rotational locomotion as well as real time place preference via optical stimulation. This technology has relevance in other contexts as well as, the alteration of gene expression at multiple time points without the need for disruptive handling of animals or surgery, with straightforward con- trol of within-subject studies.
Wireless systems for optogenetic studies of peripheral nerves
Applications of optogenetics also offer promise for dis- secting neural circuits of the spinal cord and peripheral nervous system [59,60], with additional potential rele- vance in the reduction of pain , stimulation of para- lyzed muscles  and treatment of cardiovascular dys- function . As with the brain, technologies for chronic studies depend critically on materials biocompatibility, and non-invasive mechanical interfaces with soft and highly sensitive tissues. The adaptation of conventional optical fiber approaches is particularly problematic for these applications . Here, miniaturized, wireless soft devices are highly beneficial because they provide a platform with minimal impact on soft targets .
Figure 3a shows a compact ($16mm3) lightweight ($16 mg) and stretchable (strains of up to 30%) device applied to the epidural space and the sciatic nerve in a mouse , with ability for full subdermal implantation and stable operation for 6 months or more. The technol- ogy builds on the type of device highlighted in Figure 2a. Unconditioned place aversion studies demonstrate effi- cacy for optogenetic modulation of pain. The left panel in Figure 3b shows heat maps of animals expressing ChR2 in nociceptors under the TrpV1 promoter (TrpV1-ChR2) and sensory neuron — specific gene Advillin (Advillin- ChR2), activated with wireless sciatic nerve stimulators in one arm of the Y-maze. The right-hand panel shows the same experimental setup with SNS-ChR2 mice with
Current Opinion in Neurobiology 2018, 50:1–8
Implantable, wireless device platforms for neuroscience research Gutruf and Rogers 5
Battery-free optogenetic stimulators: (a) Soft, stretchable multichannel optogenetic stimulator. Heat maps show an experimental demonstration of place aversion and preference using selective stimulation of ventral and dorsal regions of the nucleus accumbens shell . (b) Monolithically defined, thin, stretchable optogenetic stimulator for application in the deep brain. Heat maps show results of place preference experiments with stimulated Cre subjects that express ChR2 in the mesolimbic dopaminergic (DA) terminals of the nucleus accumbens and non-Cre control animals . (c) Miniaturized subdermal implant with capabilities for optogenetic stimulation of the motor cortex as shown by increased circling rates compared to a control . Wireless multimodal devices. (d) Battery powered wireless implant for in situ pharmacology. Movement traces demonstrate Mu-Opioid induced rotational locomotion and optogenetic (ChR2 expressing animals) induced place preference .
Multi-channel stretchable antenna
@ 2.3 GHz
2 mm @ 2.7 GHz
μ-ILED Injectable needle
Power receiving coil Acrylic potting
4 3 2 1 0
Low Wireless optofluidic system
Brain phantom Drug
Rectifier Circuit board
Extension Micro-LED 2 mm
20 Hz Stim
200 pmol DAMGO
DAT – Cre-
DAT – Cre+
20 Hz Stim
Current Opinion in Neurobiology
epidural implants, where a robust place aversion is also evident.
Figure 3c provides a schematic illustration of a miniatur- ized device that uses a slightly modified energy harvester to accommodate the vertical mounting orientation shown in Figure 2c with extension wires connecting to a m-ILED (volume 25 mm3, weight 50 mg) positioned adjacent to the triceps surae muscle group to allow for minimal disruption from joint rotation . The need for this careful positioning highlights a key drawback of mechanically hard devices of this type over stretchable platforms as shown in Figure 3a. Place aversion tests
involving illumination of nerve endings in the heel of the paw of ChR2 + mice show expected results, as in the right-hand panel of Figure 3c.
Advances in neuroscience often follow from the develop- ment of new tools and measurement techniques. The emerging wireless technologies summarized in this article can, through their diverse modes of operation, create significant new opportunities for research. Specifically, recent developments allow for ultraminiaturized, high- performance optoelectronic/microfluidic devices, wire- less power delivery systems and soft mechanics designs
Current Opinion in Neurobiology 2018, 50:1–8
RF ON RF OFF
(a) RF powered optogenetic stimulator for application in the epidural space and the sciatic nerve. (b) Heat maps of place preference experiments with transgenetic mice demonstrate effective wireless optostimulation in the epidural space and the sciatic nerve . (c) Schematic illustration of a miniaturized implant positioned to stimulate cutaneous nociceptors and corresponding place preference experiment results .
RF harvester LEDs
Capacitors (5 pF)
Epidural LED stimulation
Inductor (2.7 nH)
10 20 30 40
Current Opinion in Neurobiology
Current Opinion in Neurobiology 2018, 50:1–8 http://www.sciencedirect.com
% time spent in resonant cavity chamber
in platforms that can be fully implanted and operated continuously for long periods of time without adverse effects on even the smallest animal models. Active pro- grams in advanced technologies that combine sensing, stimulation and high data rate information transfer have the potential to enable closed feedback loops that can be responsive, in real time, to biological processes. The prospects are significant not only for research, but also for eventual clinical translation in therapeutic devices and bioelectronic medicines.
Conflict of interest statement
We acknowledge support from the Center for Bio-Integrated Electronics at Northwestern University and from NIH grant R21EY027612A.
References and recommended reading
Papers of particular interest, published within the period of review, have been highlighted as:
of special interest
of outstanding interest
- Boyden ES, Zhang F, Bamberg E, Nagel G, Deisseroth K: Millisecond-timescale, genetically targeted optical control of neural activity. Nat Neurosci 2005, 8:1263-1268.
- Deisseroth K: Optogenetics. Nat Methods 2011, 8:26-29.
- Stosiek C, Garaschuk O, Holthoff K, Konnerth A: In vivo two-photon calcium imaging of neuronal networks. Proc Natl Acad Sci U S A 2003, 100:7319-7324.
- Grienberger C, Konnerth A: Imaging calcium in neurons. Neuron 2012, 73:862-885.
- Otis JM, Namboodiri VM, Matan AM, Voets ES, Mohorn EP, Kosyk O, McHenry JA, Robinson JE, Resendez SL et al.: Prefrontal cortex output circuits guide reward seeking through divergent cue encoding. Nature 2017, 543:103-107.
- Yizhar O, Fenno LE, Davidson TJ, Mogri M, Deisseroth K: Optogenetics in neural systems. Neuron 2011, 71:9-34.
- Aravanis AM, Wang L-P, Zhang F, Meltzer LA, Mogri MZ, Schneider MB, Deisseroth K: An optical neural interface: in vivo control of rodent motor cortex with integrated fiberoptic and optogenetic technology. J Neural Eng 2007, 4:S143.
- Kravitz AV, Freeze BS, Parker PR, Kay K, Thwin MT, Deisseroth K, Kreitzer AC: Regulation of parkinsonian motor behaviors by optogenetic control of basal ganglia circuitry. Nature 2010, 466:622.
- Liu X, Ramirez S, Pang PT, Puryear CB, Govindarajan A, Deisseroth K, Tonegawa S: Optogenetic stimulation of a hippocampal engram activates fear memory recall. Nature 2012, 484:381-385.
- Urban DJ, Roth BL: DREADDs (designer receptors exclusively activated by designer drugs): chemogenetic tools with therapeutic utility. Annu Rev Pharmacol Toxicol 2015, 55: 399-417.
- Gilletti A, Muthuswamy J: Brain micromotion around implants in the rodent somatosensory cortex. J Neural Eng 2006, 3:189.
- Barrese JC, Rao N, Paroo K, Triebwasser C, Vargas-Irwin C, Franquemont L, Donoghue JP: Failure mode analysis of silicon- based intracortical microelectrode arrays in non-human primates. J Neural Eng 2013, 10:066014.
- Polikov VS, Tresco PA, Reichert WM: Response of brain tissue to chronically implanted neural electrodes. J Neurosci Methods 2005, 148:1-18.
- Ghosh KK, Burns LD, Cocker ED, Nimmerjahn A, Ziv Y, El Gamal A, Schnitzer MJ: Miniaturized integration of a fluorescence microscope. Nat Methods 2011, 8:871-878.
- Hui SYR, Zhong W, Lee CK: A critical review of recent progress in mid-range wireless power transfer. IEEE Trans Power Electron 2014, 29:4500-4511.
- Hwang S-W, Tao H, Kim D-H, Cheng H, Song J-K, Rill E, Brenckle MA, Panilaitis B, Won SM et al.: A physically transient form of silicon electronics. Science 2012, 337:1640-1644.
- Jeong J-W, Shin G, Park SI, Yu KJ, Xu L, Rogers JA: Soft materials in neuroengineering for hard problems in neuroscience. Neuron 2015, 86:175-186.
- Branner A, Stein RB, Fernandez E, Aoyagi Y, Normann RA: Long- term stimulation and recording with a penetrating microelectrode array in cat sciatic nerve. IEEE Trans Biomed Eng 2004, 51:146-157.
- Veiseh O, Doloff JC, Ma M, Vegas AJ, Tam HH, Bader AR, Li J, Langan E, Wyckoff J et al.: Size-and shape-dependent foreign body immune response to materials implanted in rodents and non-human primates. Nat Mater 2015, 14:643.
- Lopez CM, Andrei A, Mitra S, Welkenhuysen M, Eberle W, Bartic C, Puers R, Yazicioglu RF, Gielen GG: An implantable 455-active- electrode 52-channel CMOS neural probe. IEEE J Solid-State Circuits 2014, 49:248-261.
- Kwon KY, Lee H-M, Ghovanloo M, Weber A, Li W: A wireless slanted optrode array with integrated micro leds for optogenetics. IEEE 27th International Conference on Micro Electro Mechanical Systems (MEMS). IEEE; 2014:813-816.
- Scharf R, Tsunematsu T, McAlinden N, Dawson MD, Sakata S, Mathieson K: Depth-specific optogenetic control in vivo with a scalable, high-density mLED neural probe. Sci Rep 2016, 6:28381.
- Pisanello F, Sileo L, Oldenburg IA, Pisanello M, Martiradonna L, Assad JA, Sabatini BL, De Vittorio M: Multipoint-emitting optical fibers for spatially addressable in vivo optogenetics. Neuron 2014, 82:1245-1254.
- Gagnon-Turcotte G, LeChasseur Y, Bories C, Messaddeq Y, De Koninck Y, Gosselin B: A wireless headstage for combined optogenetics and multichannel electrophysiological recording. IEEE Trans Biomed Circuits Syst 2017, 11:1-14.
- Turcotte G, Camaro C-OD, Kisomi AA, Ameli R, Gosselin B: A wireless multichannel optogenetic headstage with on-the-fly spike detection. IEEE International Symposium on Circuits and Systems (ISCAS); IEEE: 2015:1758-1761.
- Liu J, Fu T-M, Cheng Z, Hong G, Zhou T, Jin L, Duvvuri M, Jiang Z, Kruskal P et al.: Syringe-injectable electronics. Nat Nanotechnol 2015, 10:629-636.
- Lee W, Kim D, Rivnay J, Matsuhisa N, Lonjaret T, Yokota T, Yawo H, Sekino M, Malliaras GG et al.: Integration of organic electrochemical and field-effect transistors for ultraflexible, high temporal resolution electrophysiology arrays. Adv Mater 2016, 28:9722-9728.
- Park AH, Lee SH, Lee C, Kim J, Lee HE, Paik S-B, Lee KJ, Kim D: Optogenetic mapping of functional connectivity in freely moving mice via insertable wrapping electrode array beneath the skull. ACS Nano 2016, 10:2791-2802.
- Ozden I, Wang J, Lu Y, May T, Lee J, Goo W, O’Shea DJ, Kalanithi P, Diester I et al.: A coaxial optrode as multifunction write-read probe for optogenetic studies in non-human primates. J Neurosci Methods 2013, 219:142-154.
- Seo D, Neely RM, Shen K, Singhal U, Alon E, Rabaey JM, Carmena JM, Maharbiz MM: Wireless recording in the peripheral nervous system with ultrasonic neural dust. Neuron 2016, 91:529-539.
- Canales A, Jia X, Froriep UP, Koppes RA, Tringides CM, Selvidge J, Lu C, Hou C, Wei L et al.: Multifunctional fibers for simultaneous optical, electrical and chemical interrogation of neural circuits in vivo. Nat Biotechnol 2015, 33:277-284.
Implantable, wireless device platforms for neuroscience research Gutruf and Rogers 7
Current Opinion in Neurobiology 2018, 50:1–8
- Wu F, Stark E, Ku P-C, Wise KD, Buzsa ́ ki G, Yoon E: Monolithically integrated mLEDs on silicon neural probes for high-resolution optogenetic studies in behaving animals. Neuron 2015, 88:1136-1148.
- Lee ST, Williams PA, Braine CE, Lin D-T, John SW, Irazoqui PP: A miniature, fiber-coupled, wireless, deep-brain optogenetic stimulator. IEEE Trans Neural Syst Rehabil Eng 2015, 23:655-664.
- Kim T-I, McCall JG, Jung YH, Huang X, Siuda ER, Li Y, Song J, Song YM, Pao HA et al.: Injectable, cellular-scale optoelectronics with applications for wireless optogenetics. Science 2013, 340:211-216.
- Jeong J-W, McCall JG, Shin G, Zhang Y, Al-Hasani R, Kim M, Li S,
Sim JY, Jang K-I et al.: Wireless optofluidic systems for
programmable in vivo pharmacology and optogenetics. Cell
In situ wireless pharmacology and optogenetic tool with up to four distinct drug reservoirs for programmable drug delivery to the deep brain.
- Meitl MA, Zheng-Tao Z, Kumar V, Lee KJ, Xue F, Huang YY, Adesida I, Nuzzo RG, Rogers JA: Transfer printing by kinetic control of adhesion to an elastomeric stamp. Nat Mater 2006, 5:33.
- Park S-I, Xiong Y, Kim R-H, Elvikis P, Meitl M, Kim D-H, Wu J, Yoon J, Yu C-J et al.: Printed assemblies of inorganic light- emitting diodes for deformable and semitransparent displays. Science 2009, 325:977-981.
- KimD-H,LuN,MaR,KimY-S,KimR-H,WangS,WuJ,WonSM, Tao H et al.: Epidermal Electron. Science 2011, 333:838-843.
- Gagnon-Turcotte G, Kisomi AA, Ameli R, Camaro C-OD, LeChasseur Y, Ne ́ ron J-L, Bareil PB, Fortier P, Bories C et al.: A wireless optogenetic headstage with multichannel electrophysiological recording capability. Sensors 2015, 15:22776-22797.
- Gagnon-Turcotte G, LeChasseur Y, Bories C, De Koninck Y, Gosselin B: A wireless optogenetic headstage with multichannel neural signal compression. Biomedical Circuits and Systems Conference (BioCAS); IEEE: 2015:1-4.
- Fan D, Rich D, Holtzman T, Ruther P, Dalley JW, Lopez A, Rossi MA, Barter JW, Salas-Meza D et al.: A wireless multi- channel recording system for freely behaving mice and rats. PLoS ONE 2011, 6:e22033.
- Hashimoto M, Hata A, Miyata T, Hirase H: Programmable wireless light-emitting diode stimulator for chronic stimulation of optogenetic molecules in freely moving mice. Neurophotonics 2014, 1:011002.
- Fan B, Li W: Miniaturized optogenetic neural implants: a review. Lab Chip 2015, 15:3838-3855.
- Merli F, Bolomey L, Zurcher J-F, Corradini G, Meurville E, Skrivervik AK: Design, realization and measurements of a miniature antenna for implantable wireless communication systems. IEEE Trans Antennas Propag 2011, 59:3544-3555.
- Park SI, Shin G, McCall JG, Al-Hasani R, Norris A, Xia L,
Brenner DS, Noh KN, Bang SY et al.: Stretchable multichannel
antennas in soft wireless optoelectronic implants for
optogenetics. Proc Natl Acad Sci U S A 2016, 113:E8169-E8177. Far field powered subdermal wireless optogenetic implants with multi- channel capability for multisite and multi wavelength stimulation.
- Karacolak T, Cooper R, Topsakal E: Electrical properties of rat skin and design of implantable antennas for medical wireless telemetry. IEEE Trans Antennas Propag 2009, 57:2806-2812.
- Huang X, Liu Y, Kong GW, Seo JH, Ma Y, Jang K-I, Fan JA, Mao S, Chen Q et al.: Epidermal radio frequency electronics for wireless power transfer. Microsyst Nanoeng 2016, 2:16052.
- Ho JS, Tanabe Y, Iyer SM, Christensen AJ, Grosenick L, Deisseroth K, Delp SL, Poon AS: Self-tracking energy transfer for neural stimulation in untethered mice. Phys Rev Appl 2015, 4:024001.
- Park SI, Shin G, Banks A, McCall JG, Siuda ER, Schmidt MJ, Chung HU, Noh KN, Mun JG-H et al.: Ultraminiaturized
photovoltaic and radio frequency powered optoelectronic systems for wireless optogenetics. J Neural Eng 2015, 12:056002.
- Montgomery KL, Yeh AJ, Ho JS, Tsao V, Iyer SM, Grosenick L, Ferenczi EA, Tanabe Y, Deisseroth K et al.: Wirelessly powered,
fully internal optogenetics for brain, spinal and peripheral
circuits in mice. Nat Methods 2015, 12:969-974.
Miniaturized far field powered devices for application in peripherals and brain with a hard material system.
- Kim J, Gutruf P, Chiarelli AM, Heo SY, Cho K, Xie Z, Banks A, Han S, Jang KI et al.: Miniaturized battery-free wireless systems for wearable pulse oximetry. Adv Funct Mater 2017:27.
- Jia Y, Wang Z, Mirbozorgi SA, Ghovanloo M: A closed-loop wireless homecage for optogenetic stimulation experiments. IEEE Biomedical Circuits and Systems Conference (BioCAS); IEEE: 2015:1-4.
- Kim J, Banks A, Xie Z, Heo SY, Gutruf P, Lee JW, Xu S, Jang KI, Liu F et al.: Miniaturized flexible electronic systems with wireless power and near-field communication capabilities. Adv Funct Mater 2015, 25:4761-4767.
- Cecil S, Schmid G, Lamedschwandner K, Morak J, Schreier G, Oberleitner A, Bammer M: Numerical assessment of specific absorption rate in the human body caused by NFC devices. Second International Workshop on Near Field Communication (NFC); IEEE: 2010:65-70.
- Park SI, Brenner DS, Shin G, Morgan CD, Copits BA, Chung HU, Pullen MY, Noh KN, Davidson S et al.: Soft, stretchable, fully
implantable miniaturized optoelectronic systems for wireless
optogenetics. Nat Biotechnol 2015, 33:1280-1286.
Soft far field powered subdermal wireless optogenetic implants for application in the peripheric nervous system.
- Shin G, Gomez AM, Al-Hasani R, Jeong YR, Kim J, Xie Z, Banks A, Lee SM, Han SY et al.: Flexible near-field wireless
optoelectronics as subdermal implants for broad applications
in optogenetics. Neuron 2017, 93 509–521.e503.
Subdermal wireless optogenetic implant with magnetic resonant power transfer, adaptable to most common behavioral contexts and tailored wavelength to most common in vivo opsins.
- Kim J, Banks A, Cheng H, Xie Z, Xu S, Jang KI, Lee JW, Liu Z, Gutruf P et al.: Epidermal electronics with advanced capabilities in near-field communication. Small 2015, 11: 906-912.
- Klapoetke NC, Murata Y, Kim SS, Pulver SR, Birdsey-Benson A, Cho YK, Morimoto TK, Chuong AS, Carpenter EJ et al.: Independent optical excitation of distinct neural populations. Nat Methods 2014, 11:338-346.
- Montgomery KL, Iyer SM, Christensen AJ, Deisseroth K, Delp SL: Beyond the brain: optogenetic control in the spinal cord and peripheral nervous system. Science Transl Med 2016, 8:337rv335.
- Towne C, Montgomery KL, Iyer SM, Deisseroth K, Delp SL: Optogenetic control of targeted peripheral axons in freely moving animals. PLoS ONE 2013, 8:e72691.
- Draxler P, Honsek SD, Forsthuber L, Hadschieff V, Sandku ̈ hler J: VGluT3+ primary afferents play distinct roles in mechanical and cold hypersensitivity depending on pain etiology. J Neurosci 2014, 34:12015-12028.
- Bruegmann T, Van Bremen T, Vogt CC, Send T, Fleischmann BK, Sasse P: Optogenetic control of contractile function in skeletal muscle. Nat Commun 2015:6.
- Vogt CC, Bruegmann T, Malan D, Ottersbach A, Roell W, Fleischmann BK, Sasse P: Systemic gene transfer enables optogenetic pacing of mouse hearts. Cardiovasc Res 2015, 106:338-343.
- Bonin RP, Wang F, Desrochers-Couture M, GaS secka A, Boulanger M-E, Coˆ te ́ DC, De Koninck Y: Epidural optogenetics for controlled analgesia. Mol Pain 2016, 12 1744806916629051.
Current Opinion in Neurobiology 2018, 50:1–8